When determining a disinfection plan, it is important to consider the function of the needleless connector in use. Common equipment defects include:
It is difficult to disinfect the gap between the valve and the hub
Complex contact surfaces are more difficult to disinfect
Due to the complexity of multi-part equipment, needleless connectors may hide bacteria inside the valve mechanism but outside the fluid path
Health care professionals can use many needleless connector options, and some of the above characteristics will affect the difficulty of connector sterilization.
Here are some key features to consider when evaluating pinless connectors:
- External access surface: Although some needleless connectors may seem to have a relatively flat access surface at first glance, a closer inspection reveals the different types of surfaces that need to be considered during the disinfection process. See if the surface is flat, sealed and strong. In addition, please check if there are grooves or gaps and if there are gaps around the plunger. These areas are often difficult or impossible to clean adequately, and bacteria in these spaces can cause contamination of the fluid path. Although the sterilization guidelines vary, one needleless connector has been proven to effectively remove bacteria in just 3 seconds, while other needleless connectors may require 15 seconds or more to reduce the risk of bacterial transfer. It is important to check the specific data of each connector, because the function of each connector to access the surface may affect the effective disinfection.
- Internal mechanism and channel to the fluid path: Please observe whether the internal mechanism is simple or complex, because the fluid may accidentally leak or flush into the gap, and the space between the internal mechanism and the connector housing may accumulate fluid and contaminants . The interstitial space cannot be rinsed or disinfected, and contaminants can be transferred from the interstitial space to the Luer connector and eventually to the fluid path. Internal design also has an impact on flushing, so it is important to adjust the flushing process to ensure proper flushing of the pipeline. The amount of fluid required to flush each needleless connector and catheter may vary, and clinical practice indicates that this range may be between 5ml-20ml.
- Visibility: The transparent connector enables healthcare professionals to more fully assess whether any residual blood or other infusions remain in the connector, such as blood, total parenteral nutrition or lipid emulsion. On the other hand, an opaque connector or a connector with multiple components can block the fluid path, which can increase the challenge. The clearly visible fluid path facilitates flushing of the joint, thereby clearing the catheter and helping to keep the catheter open.
- Blood backflow: Even with a small number of needles, a needleless connector that will show negative fluid displacement will cause blood backflow into the catheter when the syringe or venous infusion tube is disconnected. Conversely, if the connector does not produce a backflow, such as in the case of a true neutral displacement connector or a positive displacement connector in theory, no backflow will occur, so a specific clamping sequence may not be required. Although positive or truly neutral connectors can help catheter patency and catheter tip maintenance, the key point about return / displacement is that the fluid movement of the catheter tip is not related to the microbial invasion of the connector entry surface.
Taking into account these characteristics, a recent survey of professional clinicians found that a smooth outer surface, no gaps in the septum seals, and a transparent housing are important characteristics of a needleless connector.
With multiple choices with different key functions, the need for proper training and standardized protocols to evaluate the effectiveness of disinfection are critical to improving patient care.
Yuhuan Shengjiu Mould Co., Ltd. provides safe, fast and simple needleless injection site products suitable for medical Needleless Injection Site application. And provide customers with a complete mould configuration program to help and improve the production of disposable medical equipment. According to the requirements of the market and customers, Shengjiu has developed a new medical mould with a full hot runner system, and developed various new scientific medical equipment for customers. Welcome to consult and buy: https://www.medicalmould.net/product/